FL720/520 LED Shadowless operating light
FL720/520 LED Shadowless operating light：
operating light Used for lighting surgical, use to observe small, low-contrast objects at different depths in the incision and body control. Since the surgeon’s head, hand and instrument may interfere with the shadow of the operation, the surgery should be to eliminate the shadow and Minimize color distortion. In the surgical lamp also need to continue to work over a long period of time, but not too much heat, because the overheating that discomfort, the surgery would also can make the organization in surgical area dry. because overheating will make people ill operation, also can make the organization in surgical area dry.Shadowless operating lamp is Shared by a set of lights illuminate produced an effect, it is can increase the brightness, it is not at work to produce a shadow. The function used in the process of the doctor surgery can make doctors can more clearly see all kinds of organizations, to avoid operation error.
FL720/520 LED Shadowless operating light technique Data:
|Diameter of dome||700/500mm|
|Color temperature adjusment||3500-5000K|
|Color indication index||≧95Ra|
|L1+L2 Illumination depth||140/120cm|
|Ultraviolet light irradicance within 40mm wave legth||0.002w/m2|
|Temperature rise at surgical ´s head||≦1℃|
|Installed range(stand arm)||2800-3200mm|
|Average service life||≧50,000hrs|
|Optional||Emergency battery,camera system|
FL720/520 LED Shadowless operating light use method:See our video instructions
FL720/520 LED Shadowless operating light method:
1，Check the fasteners of LED lights regularly to prevent accidents.
2，If the LED lamp is found to be broken, the above method will be replaced. If the manufacturer is not allowed to change, notify the manufacturer for replacement.
3，Always use wet soft cloth or sponge to touch a small amount of alcohol, scrub the surface of LED lamp and the Y tube, strictly prohibit water or other liquid to wash.
4，To clean and disinfect the lamp’s handle regularly, which can be disinfected with 75% alcohol, formalin, new jal, or soas or fumigation; Not more than three days or surgery or 5 times every time before the operation, the lamp body handle coat for 138 ℃ high temperature cooking disinfection.
5，there is any other malfunction, unable to deal with, please contact the company after-sales service department, have not been evaluated by the company personnel or full-time shadowless lamp repair personnel recognition, please do not remove the leds and control circuit at will.
FL720/520 LED Shadowless operating light Product packaging：